|TUPA22||Measurements of Bunch Length in the Advanced Photon Source Booster Synchrotron||booster, synchrotron, detector, photon||394|
Funding: Work supported by the U.S. D.O.E.,Office of Science, Office of Basic Energy Sciences, under contract number DE-AC02- 06CH11357.
A bunch duration monitor (BDM) was installed at the end of a synchrotron light monitor (SLM) port in the Advanced Photon Source (APS) booster synchrotron. The BDM is based on a fast Hamamatsu metal-semiconductor-metal detector with nominal rise and fall times of 30 ps. Bunch length data is especially important as the bunch charge will be raised from 3 nC, used in the existing machine, to as much as 18 nC for APS-Upgrade operation. During preliminary high-charge studies, the SLM image is observed to move over a period of minutes while the BDM signal intensity varies; the motion is likely due to thermal loading of the in-tunnel synchrotron light mirror. Work is underway to stabilize the position using a simple feedback system and motorized mirror mount, as well as a new synchrotron light mirror assembly with improved thermal load handling. The feedback system will maintain optical alignment on the BDM at an optimum position based on the SLM centroid location. The optical layout and feedback system will be presented along with preliminary bunch length data.
|DOI •||reference for this paper ※ doi:10.18429/JACoW-NAPAC2022-TUPA22|
|About •||Received ※ 04 August 2022 — Revised ※ 09 August 2022 — Accepted ※ 10 August 2022 — Issue date ※ 09 September 2022|
|Cite •||reference for this paper using ※ BibTeX, ※ LaTeX, ※ Text/Word, ※ RIS, ※ EndNote (xml)|